A Hemostatic Device Combined with a Hemostatic Pad
Novel combination of compression device and ChitoClot Pad delivers more active and stronger coagulation effect. Continuous and stable compression can reduce a patient’s blood loss and the risk of complications, such as limb swelling, limb numbness and cold limbs. No time-consuming manual compression is needed. Smart rotating knob design enables medical staff to easily adjust pressure to a moderate level and adjust according to patient’s hemorrhage condition at any time. Soft flannel belt can be used for long duration without causing swelling, itching, irritation, or allergy.
- A novel compression device with chitosan pad for efficient artery bleeding control
- Adjustable rotating knob for point pressure and optimal patient comfort
- Transparent structure for better observation of the puncture site
- Rapid hemorrhage control to reduce the risk of complication
Silicone rubber pad
Maintains stable and uniform pressure on the surface.
Smart rotating knob
The adjustable rotating knob is easy-to-use and the pressure could be easily adjusted at any time. The intuitive design helps the medical staff to rapidly adjust the pressure according to the patient condition to avoid any patient discomfort causing by excessive pressure. Besides, the risk of postoperative complications was effectively reduced.
Soft flannel belt
Special Velcro provides a firm and comfortable fixation under tension.
Plastic release sheet
Designed to enable medical staff to first remove the compression device. The ChitoClot pad can be left on the puncture site for further processing.
The hemostatic pad effectively reduced blood loss and caused the lower chance of secondary injury contributed by adhesion to the surface of the wound.
Instruction for use
ChitoClot Artery Compression Device (Femoral)
Withdraw the sheath for 2~3 cm and clean the skin incision area.
Position the centre point of silicone pressure mat to the femoral puncture site.
After rotating the thigh from the inner side, fix the short flannel band to the black velcro.
After rotating waist from patient’s dorsal side to abdominal side, fix the long flannel band to the black velcro.
Withdraw the sheath in a parallel-to-leg direction and quickly apply pressure with the device to stop bleeding.
Keep rotating the knob clockwise to the end until there’s no bleeding from the puncture site.
*The AC-2-2 and AC-2-4 (For femoral artery) model is designed to be used for the right femoral artery only, and not suitable to be applied on the left side. Source：ChitoClot Artery Compression Device IFU
|Composition||Knob, Fixing plate, Pressure plate, Silicone pressure mat, Flannel belt, ChitoClot pad|
|Application||Bleeding management after radial intervention surgery|
|Model||AC-2-1 (with twin adhesive)
AC-2-2 (with ChitoClot Pad)